Pacing modes which allow sensing in the atrium to trigger a ventricular response are contraindicated in the presence of rapid atrial arrhythmias such as atrial fibrillation or atrial flutter. Ventricular Output Increments Range Increment 0. Page Precautions Precautions Random Failures The physician should be aware that operational failure of the Model temporary pacemaker can occur as the result of battery depletion, mishandling, or random component failure. Page Safety And Technical Checks Safety and Technical Checks Safety and technical checks should be carried out on the Model at least once every 12 months and after any malfunction or accident. Page Handling Indwelling Leads When handling indwelling leads, the terminal pins or exposed metal are not to be touched nor be allowed to contact electrically conductive or wet surfaces. This can result in a large current flowing through the implanted lead system and temporary pacemaker, which could reduce intended defibrillation energy delivered to the patient or cause myocardial damage. Page 55 RATE.
Quick Links. Table of Contents. Related Manuals for Medtronic No related manuals. Do not dispose of this product in the unsorted municipal waste Dispose of this product according to local regulations. For US audiences only Package contents Temporary pacemaker Page General Description The device operates using a 9-volt alkaline or lithium battery, installed in the battery drawer at the lower right side of the device. Note: The Model is a constant current device. When it emits a pulse, the current output is maintained at a constant value. This value is set by the output control and does not vary. When implanting a permanent pacemaker, however, Medtronic recommends the use of a Medtronic Pacing System Analyzer. Page 13 Asynchronous pacing is contraindicated in the presence of intrinsic cardiac rhythms. High-rate burst pacing in the ventricle may result in life-threatening arrhythmias.
You just suchen singlesleipzig.de a link to go to another website. If you chambeer, you will leave this site and chaber to a site medtronic single chamber temporary pacemaker by someone else. Medtronic Canada does not review or control the content on the other website, and is not responsible for any business dealings or transactions you have there.
Advisa MRI Pacemakers. External Pacemakers. Medtronic external pacemakers are intended for use with a pacemaker lead system for temporary single or dual chamber pacing in a clinical environment. Pacing Leads. The family of CapSure pacing leads for bradycardia therapy offers a choice of pacemaker leads with proven performance. Nippoldt, Doug; Whiting, Jon.
November Medtronic Data on File. Cancel Continue. Your browser is out of date With an updated browser, you will have a better Medtronic website experience. Healthcare Professional Site. By choosing to accept, er interesse hat flirten acknowledge that you are a Certified Healthcare Professional. Cancel Accept. Back to Services Integrated Health Solutions. Leads Quick Apcemaker Pacing Leads The family tempoary CapSure pacing leads for bradycardia therapy offers a choice of pacemaker leads with proven performance.
References 1 Nippoldt, Doug; Whiting, Jon. Product names on this page are trademarks or registered trademarks of Medtronic. EnRhythm Pacemaker.